I Want To Know About Acceptance Criteria Of Autoclave Validation.if I Assume Minimum F0 15 Min For Thermoucoples I Placed In The Load ,and Sal 10 -6 ,and A Kit Of B.stearothermoplillus Placed In Load.and Evaluating For Groth.it Is Enough?
hello friend its enough for indian WHO
Can any body give me the basic regulatory guidelines on autoclave validation?
is the difference between temp readings observed in between PLC & Recorder is acceptable in a sterilizer?? if yes then what is the limit??
HELLO ALL DETAIL GIVEN IN HTM 2010 FOR STERILIZAION, I AM ALSO IN INJECTION
NO +AND - 1 DEG
Please refer PDA TR-01 (2007), HTM2010 -03 & EN-285 for Autoclave Validation.
Dear Himanshu, thanks for your reply.
Can you please give me the website if possible because it will be easy for me.
The sHTM guideline in draft form you will get on internet also. or otherwise i can mail you. my mail id is email@example.com. Kindly give your email id.
My e mail ID is firstname.lastname@example.org send me the guidelines.
I have already forwarded the guideline to you. Since i am in the autoclave validation. Please free to contact me.
pl send me also that guidline sir i am also want my mail id is email@example.com sir i am working in production dept ( INJECTION)
Dear Sandeep & Vinod Kulkarni
Please check your mail , i have been send HTM 2010 on your mail.
please send the guideline on firstname.lastname@example.org i am interested to know about autoclave validation.
please send me a copy to email@example.com thanks
please send me a copy to firstname.lastname@example.org, i am interested , too.
Kind Attn : Mr. VINOD PADMAKAR KULKARNI
Request to send me regulatory guidelines on autoclave validation
My Email ID : email@example.com
Al Pharao Air Data
Please email EN285 guidline, my email id is
please send me a copy to firstname.lastname@example.org, i am also interested , too.
Form as per my suggestion HTM 2010 is the suitable guideline for autocalve validation.